WFM700 Series Waveform Monitors Service Manual 2-55
8. Move the cable to each of the remaining AES I/O connectors and repeat
step 7 for each AES I/O connector.
9. Disconnect the test setup.
Follow these steps to restore the WFM700 to the factory settings when finished
with the Performance Verification procedures.
1. Press the CONFIG button.
2. Touch the Audio Inputs/Outputs soft key and then Next Menu.
3. Touch the AES Connectors soft key, then Input.
4. Touch the Top Menu soft key.
5. Press the PRESET button.
6. Touch the -- m o r e -- soft key, then touch the Restore Factory soft key.
Restoring factory settings overrides the current instrument state.
Restore to Factory