3-16 WFM700 Series Waveform Monitors Service Manual
4. Press the CONFIG button, then touch the Calibration soft key and then the
Next Menu soft key.
5. Touch the SD Jitter Gain Adjust soft key and then the Start SD Jitter
Gain Adjust soft key.
6. Adjust the signal amplitude using the General Purpose Knob to 2 UI as
measured on the display. See Figure 3--11.
7. Touch the Save and Exit soft key to save the new calibration constant.
Figure 3- 11: SD Jitter gain adjustment screen