Bar Code Calibration Procedure without Remote Display
4. Enter the scale program mode. Power down the unit and slide the scale
program switch to the program position. If the system is a dual cable
system, disconnect the host to scale RS232 cable from the unit.
Figure 70. Entering the Scale Program Mode
5. Reinstall the platter and power up the unit.
Figure 71. Platter Installation and Power Up
6. Enter bar code calibration mode. Use the vertical window to scan the
following bar code. The scanner will beep once as it enters the bar code
calibration mode and the calibration utility will start.
Figure 72. Entering Bar Code Calibration Mode
If a razz tone is heard an error has occurred. Refer to Diagnostic
Indicator Display: Error Codes in the Stratos installation and user’s
guide for additional information (00-02983).