Scale Calibration Procedure (lbs. or kg) with Remote Display
4. Enter full service access mode. Power down the unit if necessary. Press and hold down the Calibration push button
then power up the scanner/scale. Release the Calibration push button.
Figure 41. Entering Full Service Access Mode
See Power Source caution statement located on page 11 of this manual.
5. Enter calibration mode. The remote display will flash all of the characters available (see illustration below). Press the
right arrow button twice (►)(►) while the display flashes all characters. The remote display should read CAL 1.
Figure 42. Enter Calibration Mode
6. Reinstall the platter onto the scanner/scale. Check the platter to ensure that it is seated properly and nothing is interfering
with its freedom to move.
Figure 43. Install Platter